# A-3-3-1 T8 AIN General Specs [T-Series Datasheet]

All specifications are at 25 °C and Vsupply = 5.0 V unless otherwise noted.

Parameter | Conditions | Min | Typical | Max | Units |
---|---|---|---|---|---|

Typical Input Range | Gain=0.125 | - | ±11.0 | - | Volts |

Gain=0.25 | - | ±9.6 | - | Volts | |

Gain=0.5 | - | ±4.8 | - | Volts | |

Gain=1 | - | ±2.4 | - | Volts | |

Gain=2 | - | ±1.2 | - | Volts | |

Gain=4 | - | ±0.6 | - | Volts | |

Gain=8 | - | ±0.3 | - | Volts | |

Gain=16 | - | ±0.15 | - | Volts | |

Gain=32 | - | ±0.075 | - | Volts | |

Gain=64 | - | ±0.036 | - | Volts | |

Gain=128 | - | ±0.018 | - | Volts | |

Max AIN Voltage | No Damage | -16 | - | 16 | Volts |

Input Bias Current _{[1]} | 10kΩ Load | - | 4 | - | nA |

Input Impedance _{[1]} | - | - | 1 | - | GΩ |

Max Source Impedance _{[1]} | - | - | 10 | - | kΩ |

Integral Linearity Error | Range=11V _{[2]} | - | - | ±0.003 | %FS |

Range=9.6V | - | - | ±0.005 | %FS | |

Range=4.8V | - | - | ±0.005 | %FS | |

Range=2.4V | - | - | ±0.005 | %FS | |

Range=1.2V | - | - | ±0.005 | %FS | |

Range=0.6V | - | - | ±0.007 | %FS | |

Range=0.3V | - | - | ±0.015 | %FS | |

Range=0.15V | - | - | ±0.015 | %FS | |

Range=0.075V | - | - | ±0.015 | %FS | |

Range=0.036V | - | - | ±0.050 | %FS | |

Range=0.018V | - | - | ±0.260 | %FS | |

Absolute Accuracy | Range=11V | - | 0.005 | %FS | |

Range=9.6V | - | 0.005 | %FS | ||

Range=4.8V | - | 0.006 | %FS | ||

Range=2.4V | - | 0.006 | %FS | ||

Range=1.2V | - | 0.006 | %FS | ||

Range=0.6V | - | 0.008 | %FS | ||

Range=0.3V | - | 0.012 | %FS | ||

Range=0.15V | - | 0.012 | %FS | ||

Range=0.075V | - | 0.024 | %FS | ||

Range=0.036V | - | 0.150 | %FS | ||

Range=0.018V | - | 0.320 | %FS | ||

Temperature Coefficient _{[3]} | - | - | 0.75 | - | ppm/°C |

Noise (Peak-To-Peak) _{[4]} | See A-3-3-2 | - | - | 65 | μV |

Effective Resolution (RMS) _{[4]} | See A-3-3-2 | - | - | 21.1 | bits |

Noise-Free Resolution _{[4]} | See A-3-3-2 | - | - | 18.1 | bits |

[1] The key specification here is the maximum source impedance. As long as your source impedance is not over this value, there will be no substantial errors due to impedance problems. For source impedance greater than this value, more settling time might be needed.

[2] 11V range characterized from -10 to +10 V.

[3] Tested using the 9.6V range.

[4] Tested using the 9.6V range running at 100 Hz.

See also: T8 Noise and Resolution